|Place of Origin:
|Minimum Order Quantity:
|standard exporting package
|L/C, D/A, D/P, T/T, Western Union, MoneyGram
|300 sets per month
|Blackboard Thermometer Measurement Range:
|30~80°C/tolerance Is ±1°C
|0.1 ° C
|Standard Test Piece Size:
|Good Quality Assurance:
|UV Aging Tester
Environmental UV Light Accelerated Aging Test Chamber UV Weathering Tester for Rubber, Plastic, Leather and Cloth
The light source uses 8 imported UV fluorescent lamps with a rated power of 40W as the light source. Ultraviolet fluorescent tubes are distributed on both sides of the machine, 4 on each side. There are UVA-340 and UVB-313 light sources for users to choose from. The luminescence spectrum energy of the UVA-340 lamp is mainly concentrated at the wavelength of 340 nm, and the luminescence spectrum of the UVB-313 lamp is mainly concentrated near the wavelength of 313 nm. We use the UVA-340 lamp because the fluorescent light energy output will gradually decay with time. In order to reduce the impact of the test due to the attenuation of light energy, this test box is every 1/1 of all eight lamps. When the fluorescent lamp of 4 is used, an old lamp is replaced by a new lamp, so that the ultraviolet light source is always composed of a new lamp and an old lamp, thereby obtaining a constant output light energy. The effective life of the lamp can be around 1600 hours.
1.Blackboard temperature and condensing temperature are controlled by controller.
2.The rest basically use imported electronic components.
Uniformity of irradiance: ≤ 4% (at the surface of the sample)
Blackboard temperature monitoring: The standard Pt-100 blackboard temperature sensor is used to precisely control the sample surface temperature during the test.
Blackboard temperature setting range: BPT 40-75 °C; but the actual maximum temperature limit of the internal temperature protection device is 93 °C ± 10%.
Blackboard temperature control accuracy: ±0.5 °C
3.Sink temperature monitoring: During the loop test, one test section is a dark condensation process, which requires a saturated water vapor that can generate a higher temperature in the tank. When the water vapor encounters a relatively cold surface of the sample, Dew will condense on the surface of the sample.
The sink is located in the lower part of the cabinet and has an electric heater.
Sink temperature control range: 40~60°C
4.The test box is equipped with a time controller, the range is 0~530H, power failure memory function.
5.Safety protection device:
Over temperature protection inside the box: When the temperature inside the box exceeds 93 °C ± 10%, the machine will automatically cut off the power supply of the lamp and heater, and enter the equilibrium state to cool.
The low water level alarm in the sink prevents the heater from being burned.
Contact Person: Mr. zhong