|Place of Origin:||Guangdong|
|Minimum Order Quantity:||one set|
|Packaging Details:||standard exporting wooden package|
|Delivery Time:||3 days|
|Payment Terms:||L/C, D/A, D/P, T/T, Western Union, MoneyGram|
|Supply Ability:||300 sets per month|
|Product Name:||No Rotor Rheometer For Rubber||Setting Range:||0 To 200°C|
|Setting Minimum Unit:||0.1°C||Temperature Display Range:||0 To 200°C|
|Testing Before Delivery:||Yes||After-sale Customer Service:||Yes|
Computer Control Rubber No Rotor Rheometer Testing Machine / Equipment / Instrument / Apparatus / Device / Method
Rubber`Rheometer is applied to determine the characteristics of vulcanized rubber,widely used in rubber processing industry, rubber quality control and basic research rubber.
Data acquisition system stores all data for the support of computer date base and relative software and deals with the parameter optimally.
According to the prioritization scheme, it can sort out all dates and find out the best vulcanizing parameter. Besides printing vulcanizing curve at real time, it also can show on the display and print vulcanizing curve and parameter at any time.
The test instrument has the following characteristics
a. High accuracy temperature control and wide range. (±0.1°C)
b. Clock programming function. (setting, modification time)
e. Meet the requirements of GB/T16584 (Test Rubber Vulcanization Characteristics with Rotorless Vulcanometer) and ISO6502.
Powerful microprocessors use high quality imported chips. On the one hand, computers and microprocessors can take signals from the torque sensor and record the signals on the graph. At the same time, the computer and the measurement control module interface exchange data and control the temperature of the mold cavity.
The rotorless vulcanometer was designed with a series of interlocking functions, including closing the plastic glass door. If the door is not closed, the mold cannot be closed. Only when the door is closed safely can the mold be closed. Only work can be done.
All in all, this type of vulcanizer is more stable, more accurate, cheaper, and more functional than other models. It is the best instrument for testing rubber properties.
The standard for the implementation of the vulcanization instrument
|ASTM D5289-95 ISO6502-1999(E) GB/T16584-1996|
|Temperature range||room temperature to 200 °C|
|Temperature measurement accuracy||100 to 200°C, ≤±0.1°C|
|Temperature stability||100 to 200°C, ≤±0.1°C|
|Setting range||0 to 200°C|
|Setting minimum unit||0.1°C|
|Temperature display range||0 to 200°C|
|Set the time range||0 to 200 minutes|
|Set the minimum unit||1 minute|
|Automatic range switching||automatic increase according to actual torque|
|Power||50HZ, ~220V±10%, 50±1Hz, must be well grounded|
|Compressed air||0.32Mpa controlled by pneumatic control valve|
|Swing frequency||100 times/minute (about 1.7HZ)|
|Ambient temperature||normal temperature -40°C, relative humidity less than 80%, no corrosive gas in the air|
Contact Person: Mr. zhong